Siemens Healthineers is looking for a solution to equip surfaces of large cover parts (2m²) either with a color neutral or white electrostatic discharging (ESD) surface.
Standard static dissipative surface coatings and/or materials exhibit an inherent color which is not white. In the requested case the preferred color is the color white (eg. RAL 9010, RAL 9003).
Key success Criteria
We are seeking partners that can provide a solution meeting the following criteria:
Electric surface resistance 105 ≤ R ≤ 1011Ω
Surface coating shall be either colored or a clearcoat:
Color: RAL 9003 or RAL9010
Clear coat shall not impair impression of white color (eg. RAL 9010, RAL9003)
Gloss level: glossy, semi-matt,matt
No fading of the ESD-effectiveness overtime
Resistance to active agent groups: Aldehydes, alkylamines, quaternary compounds, pyridine derivatives, guanidine derivatives, chlorine derivatives, peroxide compounds, organic acids, phenol derivatives, alcohols, benzene, common household dishwashing liquids - detergentsubstances
As scratch resistant as possible: No visible scratches at minimum 4 N with the tip vertically on the surface (hardness tester type 318Erichsen)
Resistance to irradiation: SUN test acc. to the following description: The exposure takes place for three hours without a filter (uncoated plastics for 30 hours) in a wavelength of 270nm-800nm with an intensity of 765W/m² ± 10%. The permitted color change compared with the surface that is not irradiated is dE* ≤ 1.0 (+0.2 measuring instrument tolerance).
Resistance to UV C radiation acc. to the following description: The test for paints as well as uncoated plastics takes place during 3-hour radiation with short-wave UV light (wavelength around 250nm) and an intensity of 21 W/m² ± 10%. The permitted color change compared with the surface that is not irradiated is dE* ≤ 1.0 (+0.2 measuring instrumenttolerance).
Adhesion of the coating on the substrate acc. to ISO 2409 (Gt ≤1)
Biocompatible acc. to ISO 10993-1: Categorisation acc. to Table A1, potential contact areas categoryA
Attachment of self-adhesive labels on the surface shall bepossible
The solution shall be applicable to large plastic parts (2m²). The plastic material may be ABS, PC, PS, PUR. The solution must be marketable worldwide.
The solution shall be applicable to series production: 100 to 1000 pieces per year. Approximately 5 m² - 10 m² per piece.
Siemens Healthineers is open to all solutions that meet the key success criteria and are applicable to plastic parts.
Sep 21, 2018
Items to be submitted
We are looking for concise, non-confidential proposals. The proposal should:
Describe the technical approach
Provide information on the performance of the technology (if available)
Include a description of the responding organisation or team and their experience.
To Respond, please use the Respond button located at the end of this page.
Please note that only non-confidential information can be accepted. By submitting a response you represent that the response does not contain and will not be deemed to contain any confidential information of any kind whatsoever. By submitting a response, you acknowledge that Siemens, in its sole discretion, may select or reject a response or any portion thereof. Your NineSights community profile should contain all required information in order to provide us with appropriate contact information for your responses.
For questions about NineSights privacy and security, please feel free to post in our Community Help forums.
Preferred Collaboration Types
Area of Interest
Physics-Plasma > Gases, Plasmas, Electric Discharges
Materials Science > Coatings
Materials Science > Materials Manufacturing
Materials Science > Nanomaterials
Physics-Condensed Matter > Surfaces and Interfaces, Thin Films, Nanosystems
Engineering-Chemical > Electrochemistry
Nanoscience and Nanotechnology > Surface Science
Engineering-Electrical > Electrical and Electronics